Note: Support for Nox W7 Link is discontinued in Noxturnal versions 6.0 and up.
The Radiometer has a serial output on the back. With a serial cable provided by Radiometer it is possible to connect a W7 Link - R connector to send the serial data via Bluetooth® technology to the Nox T3 recorder.
The setup image for the Radiometer can be seen below:
The following items are needed in order to setup the connection:
- Radiometer and a serial cable (Provided by Radiometer)
- Nox W7 Link-R Kit (Nox Ref: 54 40 11) which contains:
- Nox W7 Link
- Null Modem (Gender Changer)
- Nox W7 Power adapter with a Nox mini USB cable
When you configure the Nox T3 ambulatory recording you need to make sure that the Radiometer is defined in the recording type. If the setup is correct the configure device window should look like this:
Fill in the PIN number for the Nonin 3150 oximeter and serial number (SN) for the W7 Link - R (PIN and SN is found on the back of both devices). Finish the device configuration and the device should be ready to record.
For further information please contact firstname.lastname@example.org