The ResMed S9 has a serial output on the back. It is possible to connect a W7 Link - R connector to send the serial data via Bluetooth® technology to the Nox T3 recorder.
The setup image for the ResMed S9 can be seen below:
The following items are needed in order to setup the connection:
- ResMed S9 and an Embletta Adapter
- Nox W7 Link-R Kit (Nox Ref: 54 40 11) which contains:
- Nox W7 Link
- Nox W7 Power adapter with a Nox mini USB cable
When you configure the Nox T3 ambulatory recording you need to make sure that the ResMed S9 is defined in the recording type. If the setup is correct the configure device window should look like this:
Fill in the PIN number for the Nonin 3150 oximeter and the Serial number (SN) for W7 Link - R (PIN and SN is found on the back of both devices). Finish the device configuration and the device should be ready to record.
Signals available when using the Nox W7 Link - R:
- Tidal Volume
- Set Pressure
- Mask Pressure
For further information please contact firstname.lastname@example.org