Learn how to set up the Nox T3 and the Noxturnal software for T3 studies. The following article is a quick set up guide and here you will find references to the relative material for the Nox T3 and Noxturnal.
Step 1. Installation
Start by installing the Noxturnal Software. Before installing the software, review the system requirements for running the application. Instructions on how to install the software and system requirements can be found here, in the Noxturnal Software Manual. Note that you have to have administrator privilege on your computer in order to install the software.
Step 2. Start Noxturnal
When you have finished the install, you can start Noxturnal and connect the Nox T3 device with a USB cable to the computer. To connect the Nox T3 device to a computer you need to access the USB connector on the device. The USB connector is placed under the battery lid. To open the battery lid, press with the lid key (accompanying the system) and slide the battery lid down, towards the bottom of the device. Connect the device to a computer with the USB cable.
Step 3. Connect Device
Noxturnal automatically detects the device within 2-4 seconds. If the device is not detected, click Scan
When Noxturnal detects the connected device the following information about the device is displayed: recording status, firmware version and device name.
Step 5. Configure Device
Chose Prepare a New Recording
A device configuration wizard will appear. The configuration in Noxturnal is done in three steps.
Here you can chose the applicable recording template. The recording templates have descriptive names to exhibit the recording type they are used for. You can edit recording templates by pressing the edit link. The templates can also be accessed from Devices - Recording Templates.
See further instructions here.
Also, type in the applicable oximeter PIN address, then click Next to enter the next wizard:
2. Schedule Recording
Here you select, if it's a manually started recording or a automatic scheduled recording. Also, select the duration time for the recording. Then click Next:
3. Patient Information
Here you enter the patient information.
The devices now have to be paired before they are sent out. Dependent on if you have configured for a manual recording or a scheduled recording, see instructions here: