The Video device has a LAN output that is also the power supply of the device. Therefore you need a Power over ethernet (PoE) switch to power the device. It is possible to connect to the Nox Sleep System by connecting the Video LAN output to the same local network as the Nox C1 Access Point is connected to.
The setup image for the Video device can be seen below:
The following items are needed for the Nox Sleep System setup with a Video device
- Video Device (Recommended video devices)
- Nox C1 Access Point and a power supply*
- 3x Ethernet Cables
- PoE Switch
- PC Computer**
Instructions on how to set up an online room in the Noxturnal Software and how to start an online recording be found here.
Instructions on how to add auxiliary devices into online rooms in Noxturnal can be found here.
*Warning: Only use power supply FRIWO MP115 Medical-7555M/12 with the C1 device. The use of an incorrect power supply may result in electric shock or cause the device to overheat, which may result in patient/operator harm.
** Warning: The computer used must comply with the international standard IEC 60950-1 for the safety of information technology equipment.
For further information please contact email@example.com